- Research article
- Open Access
Detection of rabbit IgG by using functional magnetic particles and an enzyme-conjugated antibody with a homemade magnetic microplate
© Tsai et al.; licensee Springer. 2015
Received: 16 October 2014
Accepted: 6 February 2015
Published: 22 February 2015
The enzyme-linked immunosorbent assay (ELISA) has been used for diagnosing medical and plant pathologies. In addition, it is used for quality-control evaluations in various industries. The ELISA is the simplest method for obtaining excellent results; however, it is time consuming because the immunoreagents interact only on the contact surfaces. Antibody-labeled magnetic particles can be dispersed in a solution to yield a pseudohomogeneous reaction with antigens which improved the efficiency of immunoreaction, and can be easily separated from the unreactive substances by applying a magnetic force. We used a homemade magnetic microplate, functional magnetic particles (MPs) and enzyme-labeled secondary antibody to perform the sandwich ELISA successfully.
Using antibody-labeled MPs enabled reducing the analysis time to one-third of that required in using a conventional ELISA. The secondary antibody conjugated with horseradish peroxidase (HRP) was affinity-bound to the analyte (IgG in this study). The calibration curve was established according to the measured absorbance of the 3, 3′, 5, 5′-tetramethybezidine–HRP reaction products versus the concentrations of standard IgG. The linear range of IgG detection was 114 ng/mL–3.5 ng/mL. The limit of detection (LOD) of IgG was 3.4 ng/mL. The recovery and coefficient of variation were 100% (±7%) and 116% (±4%) for the spiked concentrations of 56.8 ng/mL and 14.2 ng/mL, respectively.
Pseudohomogeneous reactions can be performed using functional MPs and a magnetic microplate. Using antibody-labeled MPs, the analysis time can be reduced to one-third of that required in using a conventional ELISA. The substrate–enzyme reaction products can be easily transferred to another microplate, and their absorbance can be measured without interference by light scattering caused by magnetic microbeads. This method demonstrates great potential for detecting other biomarkers and in biochemical applications.
In recent years, functionalized magnetic particles have attracted considerable research interest for many biological applications such as biomedicine [1-3], isolation of specific DNA , and manipulation of cells [5,6]. In addition, MPs have been used for detecting biomarkers by using superconducting quantum interference device (SQUID) [7-9] and magnetic resonance . We used MPs for detecting biomarkers by using various techniques [11-17], such as (1) predeposited the functionalized MPs in the thin channels coupling with a magnet, then captured the biomarker antibodies and analyzed by counting particles or off-line measurement of fluorescent intensity. (2) detected biomarker antibodies using functional nanomagnetic and fluorescent nanoparticles in magnetic microplate.
A sandwich enzyme-linked immunosorbent assay (ELISA) is a frequently employed bioanalytical assay that involves using an antibody-labeled solid phase to detect the presence of a substance, generally an antigen, in a liquid sample. An enzyme-conjugated secondary antibody is then added to form a sandwich structure. Thus, the enzyme-catalyzed substrate reaction increases the sensitivity of the immunoassay. The ELISA has been used for diagnosing medical and plant pathologies. In addition, it has been used for quality-control evaluations in various industries. The ELISA is the simplest method for obtaining excellent results; however, it is time consuming because the immunoreagents interact only on the contact surfaces. We have improved the efficiency of the antigen–antibody reaction by integrating the sandwich immunoassay using functional magnetic and fluorescent nanoparticles in magnetic microplate. The magnetic microparticles were not suitable for direct measurement by microplate reader due the light scatting of the microparticles. In this study, we attempted to overcome this limitation using functional magnetic microparticles and an enzyme-conjugated antibody in a magnetic microplate. This method has many potential advantages which were reported in previous studies , such as (1) the amount of proteins immobilized on the particles is consistent in the same batch, which can be used for performing several reactions. (2) The magnetic microparticles (MPs) with avidin, carboxyl, or amino functional groups are commercial available which made antibody labeling easily. (3) MPs can be dispersed in a solution to yield a pseudohomogeneous reaction with antigens and can be easily separated from the unreactive substances by applying a magnetic force. (4) They can be redispersed in the solution after removing the magnetic force. (5) The enzyme-conjugated antibody can react with substrates pseudohomogeneously, and the products can be easily transferred from one microplate to another. The absorbance of the products can be measured without interference by light scattering caused by magnetic microbeads. The homogenous immunoreactions are more efficient than that of reaction on the surface of microplates . Thus, in using antibody-labeled MPs, the time required for analysis is expected to be less than that required by a conventional ELISA. Most literatures on magnetic particle-based ELISA were processed in tubes [19,20]. The washing steps were done with one tube by one tube or with commercial magnetic separators, such as fully automated multisampling separators. The automated multisampling separators are expensive. Thus, we fabricated a practical and inexpensive magnetic microplate. The important contribution of our current work is the integration of microplate ELISA with homemade magnetic microplate. The process of the microplate ELISA will be more easily adopted in the clinical laboratory than tube-ELISA and the home made magnetic microplate is inexpensive.
Rabbits are among the most commonly used experimental animals in the areas of biochemical research and medical products. The serum immunoglobulin levels of an animal reflect its immune status. One of our coworkers used rabbits for performing immune experiments. Therefore, we used rabbit IgG as a model analyte to demonstrate our detection method. In this study, MPs were labeled with anti-IgG, then IgG from the sample was bound to anti-IgG-MPs. A secondary antibody conjugated with horseradish peroxidase (HRP) was then used to bind to IgG. In the final step, enzyme substrates were added. The subsequent reaction produced a color change, and the absorbance of the product was measured.
Materials and methods
Chemicals and materials
An affinity isolated antibody, a buffered aqueous solution of biotinlyated antirabbit IgG antibody (whole-molecule), was produced in goat; rabbit IgG purified from a normal rabbit serum by using fractionation and ion-exchange chromatography; phosphate-buffered saline (PBS); dimethyl sulfoxide (DMSO); and 3,3′,5,5′-tetramethybezidine (TMB) were purchased from Sigma-Aldrich (Saint Louis, MO, USA). Dynabeads® MyOne™ Streptavidin T1 (streptavidin-coupled superparamagnetic beads 1 μm in diameter) and Novex® HRP-conjugated goat antirabbit IgG (H & L) antibody were purchased from Life Technologies (Grand Island, NY, USA). Triton X-100 was purchased from Tedia (Fairfield, OH, USA). Hydrogen peroxide (H2O2, 35%) was purchased from Shimakyu’s Pure Chemicals (Osaka, Japan).
Magnetic separator–magnetic microplate
Permanent magnets 6 mm in diameter and 13 mm in length were fixed in the wells of a microplate, and the assembled magnetic microplate (Additional file 1: Figure S1) was then placed under another microplate to form a magnetic separator. The magnetic field strength of these magnets was 4.1 (±0.2) kG at the top of separator.
A spectrometer (Flexstation 3 multimode plate reader, Molecular Devices, Sunnyvale, CA, USA) was used to measure the optical intensity.
Functional magnetic particle preparation (anti-IgG-labeled MPs)
Streptavidin-coupled dynabeads were conjugated with biotinylated anti-IgG, based on the extremely high binding affinity of the streptavidin–biotin interaction (Kd = 10−15), and further used for developing the pseudohomogeneous immunoassay. An aliquot of 100 μL of biotinylated anti-IgG (3.3 mg/mL) was added to a centrifuge tube containing 10 mL of PBS and 10 mg dynabeads, and the tube was then gently rotated using a MACSmix™ tube rotator for 2 h at 4°C. Anti-IgG-labeled MPs were attracted by the magnets, and the MPs were washed three times with PBS to remove unreacted anti-IgG. Finally, anti-IgG-labeled MPs were reconstituted with 10 mL of PBS and divided into aliquots of 1 mL each. The aliquots were maintained at 4°C until use. The suspensions of the unreacted and washed PBS were mixed, and the protein concentration was evaluated using a Bradford reagent. The amount of labeled anti-IgG was approximately 18 μg/mg of dynabeads, which was semiquantitative based on the added amount subtracted from the amount left in the suspension. This result was consistent with that claimed by the supplier (biotinylated IgG up to 20 μg/mg of dynabeads).
Procedures for the magnetic sandwich immunoassay
A calibration curve of the absorbance was established by plotting the measured absorbance versus the various known concentrations of IgG.
Preparation of serum and spiked samples
We drew blood from rabbits’ ears. The rabbit antisera was precipitated using (NH4)2SO4 to a final concentration of 50% and 35% in sequence by 100% saturated (NH4)2SO4 solution. The precipitate was redissolved in distilled water equal to half of the original volume and then dialyzed against 2 L PBS for 72 h at 4°C with two changes of buffer. Finally, 0.01 M PBS was added to the original volume. The rabbits were New Zealand white rabbits (body weight, 3.5 kg), which were obtained from Deer-Ho farm (Taichung, Taiwan) for immune experiments conducted by our coworkers. The animal experiments in our study were approved by the Institutional Animal Care and Use Committee at the Chung Shan Medical University (approval no. 1269). Prior to using the ELISA, the serum was diluted 10,000 times using PBS. To demonstrate the practicality of the proposed magnetic ELISA, IgG concentration in spiked serum samples was measured.
Results and discussion
Optimization of immunoreaction time
Figure 2c shows the effect of wash times on the cleanliness of the unreactive anti-IgG-HRP. We transferred the suspension of the secondary immunoreaction and washed buffers to another microplate, and then removed 20 μL of each solution to react with 200 μL of PBS containing TMB/H2O2. The results from the twice-washed buffer were not different from those of the blank that contained only the TMB/H2O2 in buffer. Therefore, we washed the microplate only twice after each immunoreaction step.
Effect of amount of magnetic beads
Regarding the cost of the dynabeads and particle aggregation, we performed dose-dependent measurements of IgG with 10 μg and 20 μg of anti-IgG-labeled MPs.
Calibration curve and reproducibility
Data of analytical specifications
Amount of anti-IgG-MPs
56.8 ~ 3.54
2 ~ 10%
113.6 ~ 3.54
1 ~ 9%
The importance of precision has often been emphasized for using a bead-based immunoassay in quantitative analysis. The intraassay precision of the analytical method was calculated by analyzing each concentration in triplicate per run. The relative standard deviation was 1%–5% at varied concentration levels, except for 9%–10% at 3.5 ng/mL. These results implied that the proposed method exhibited satisfactory reproducibility. The bioactivity of highly diluted HRP decayed rapidly. Aliquots of anti-IgG-HRP diluted 100 times were maintained at 4°C. The working solution of anti-IgG-HRP was adjusted to 500–1000 dilutions based on a positive control test of the anti-IgG-HRP–TMB/H2O2 reaction, which maintained an absorbance at 1.5. Thus, reproducibilities of the interassay were less than 7%.
Determination of IgG in rabbit serum sample
An aliquot of 220 μL of dilute rabbit serum was incubated with 20 μg of anti-IgG-labeled MPs. The recovery was measured for the spiked IgG in serum at final concentrations of 56.8 ng/mL and 14.2 ng/mL. The IgG concentration in the obtained serum and spiked solutions was measured and interpreted according to the calibration curve. The IgG concentration of the rabbit serum was 7.66 mg/mL (±3%), which is consistent with that reported (5–10 mg/mL) by a previous study . The recovery and coefficient of variation were 100% (±7%) and 116% (±4%) for the spiked concentrations of 56.8 ng/mL and 14.2 ng/mL, respectively.
We developed an ELISA that combines the sandwich immunoassay with MPs and an enzyme-conjugated secondary antibody on a magnetic microplate for determining the IgG concentration in a buffer solution and serum. The high sensitivity of the assay was achieved using the colorimetric method for measuring the activity of the conjugated HRP. The dynamic working range was 114-3.5 ng/mL. The recovery ranged from 100% to 116%, and reproducibility ranged from 1% to 10%. In using antibody-labeled MPs, the time required for analysis was reduced to one-third of that required in using a conventional ELISA. The detection limit was 3.4 ng/mL (i.e. 2.3 × 10−11 M) which was lower than 10−9–10−10 M suggested by the vendors of conventional ELISA kits and time-resolved fluorescence . The homemade magnetic microplate was practical and inexpensive. This method has satisfactory potential for detecting other biomarkers and in biochemical applications.
The authors gratefully acknowledge the financial support from the Ministry of Science and Technology, Taiwan (NSC 102-2113-M-040-002).
- Morimoto Y, Natsume H. Targeting technology utilizing magnetic microparticulate system for cancer therapy. Jpn J Clin Med. 1998;56:649–56.Google Scholar
- Saini G, Shenoy D, Nagesha DK, Kautz R, Sridhar S, Amiji M. Superparamagnetic iron oxide-gold core-shell nanoparticles for biomedical applications. 2005 NSTI Nanotechnology Conference and Trade Show - NSTI Nanotech 2005 Technical Proceedings. 2005;328-331.Google Scholar
- Blackburn WH, Dickerson EB, Smith MH, McDonald JF, Lyon LA. Peptide-functionalized nanogels for targeted siRNA delivery. Bioconjug Chem. 2009;20:960–8.View ArticleGoogle Scholar
- Kinsella JM, Ivanisevic A. Enzymatic clipping of DNA wires coated with magnetic nanoparticles. J Am Chem Soc. 2005;127:3276–7.View ArticleGoogle Scholar
- Pisanic Ii TR, Blackwell JD, Shubayev VI, Fiñones RR, Jin S. Nanotoxicity of iron oxide nanoparticle internalization in growing neurons. Biomaterials. 2007;28:2572–81.View ArticleGoogle Scholar
- Van Den Bos EJ, Wagner A, Mahrholdt H, Thompson RB, Morimoto Y, Sutton BS, et al. Improved Efficacy of Stem Cell Labeling for Magnetic Resonance Imaging Studies by the Use of Cationic Liposomes. Cell Transplant. 2003;12:743–56.View ArticleGoogle Scholar
- Horng HEY SY, Huang YW, Jiang WQ, Hong C-Y, Yang HC. Nanomagnetic particles for SQUID-based magnetically labeled immunoassay. IEEE Trans Appl Supercond. 2005;15:668–71.View ArticleGoogle Scholar
- Tsukamoto A, Kuma H, Saitoh K, Kandori A, Yoshinaga K, Sugiura Y, et al. Reduction of the magnetic signal from unbound magnetic markers for magnetic immunoassay without bound/free separation. Physica C. 2007;463–465:1024–8.View ArticleGoogle Scholar
- Kuma H, Oyamada H, Tsukamoto A, Mizoguchi T, Kandori A, Sugiura Y, et al. Liquid phase immunoassays utilizing magnetic markers and SQUID magnetometer. Clin Chem Lab Med. 2010;48:1263–9.View ArticleGoogle Scholar
- Wagner S, Schnorr J, Pilgrimm H, Hamm B, Taupitz M. Monomer-coated very small superparamagnetic iron oxide particles as contrast medium for magnetic resonance imaging: Preclinical in vivo characterization. Invest Radiol. 2002;37:167–77.View ArticleGoogle Scholar
- Fuh CB, Su YS, Tsai HY. Determination of magnetic susceptibility of various ion-labeled red blood cells by means of analytical magnetapheresis. J Chromatogr A. 2004;1027:289–96.View ArticleGoogle Scholar
- Tsai HY, Yin C, Lin YP, Fuh CB. New method of blood typing using analytical magnetapheresis. J Chromatogr A. 2006;1120:35–7.View ArticleGoogle Scholar
- Tsai HY, Hsu CF, Chiu IW, Fuh CB. Detection of C-reactive protein based on immunoassay using antibody-conjugated magnetic nanoparticles. Anal Chem. 2007;79:8416–9.View ArticleGoogle Scholar
- Tsai H, Lin Y, Chang HW, Fuh CB. Integrating the QCM detection with magnetic separation for on-line analysis. Biosens Bioelectron. 2008;24:485–8.View ArticleGoogle Scholar
- Tsai HY, Jian SJ, Huang ST, Fuh CB. Competitive magnetic immunoassay for protein detection in thin channels. J Chromatogr A. 2009;1216:7493–6.View ArticleGoogle Scholar
- Tsai HY, Chan JR, Li YC, Cheng FC, Fuh CB. Determination of hepatitis B surface antigen using magnetic immunoassays in a thin channel. Biosens Bioelectron. 2010;25:2701–5.View ArticleGoogle Scholar
- Tsai HY, Chang CY, Li YC, Chu WC, Viswanathan K, Bor Fuh C. Detection of carcinoembryonic antigen using functional magnetic and fluorescent nanoparticles in magnetic separators. J Nanopart Res. 2011;13:2461–7.View ArticleGoogle Scholar
- Farrell S, Ronkainen-Matsuno NJ, Halsall HB, Heineman WR. Bead-based immunoassays with microelectrode detection. Anal Bioanal Chem. 2004;379:358–67.View ArticleGoogle Scholar
- Song F, Zhou Y, Li YS, Meng XM, Meng XY, Liu JQ, et al. A rapid immunomagnetic beads-based immunoassay for the detection of β-casein in bovine milk. Food Chem. 2014;158:445–8.View ArticleGoogle Scholar
- Lei JH, Guan F, Xu H, Chen L, Su BT, Zhou Y, et al. Application of an immunomagnetic bead ELISA based on IgY for detection of circulating antigen in urine of mice infected with Schistosoma japonicum. Vet Parasitol. 2012;187:196–202.View ArticleGoogle Scholar
- Li H, Chen J, Xu F, Cao H, Maria E. Optimization of TMB substrate chromogenic system in ELISA and the study of its stability in storage. Biotechnol Bull. 2010;2:126–30.Google Scholar
- Farrell Jr HM, Jimenez-Flores R, Bleck GT, Brown EM, Butler JE, Creamer LK, et al. Nomenclature of the proteins of Cows’ milk—sixth revision. J Dairy Sci. 2004;87:1641–74.View ArticleGoogle Scholar
- Ozinskas AJ. Topics in Fluorescence Spectroscopy: Volume 4: Probe Design and Chemical Sensing. In: Springer Science & Business Media. 1994.Google Scholar
This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly credited. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.