- Research article
- Open Access
Electrochemical DNA biosensor based on the BDD nanograss array electrode
© Jin et al.; licensee Chemistry Central Ltd. 2013
Received: 14 February 2013
Accepted: 3 April 2013
Published: 10 April 2013
The development of DNA biosensor has attracted considerable attention due to their potential applications, including gene analysis, clinical diagnostics, forensic study and more medical applications. Using electroactive daunomycin as an indicator, the hybridization detection was measured by differential pulse voltammetry in this study.
Electrochemical DNA biosensor was developed based on the BDD film electrode (fBDD) and BDD nanograss array electrode (nBDD). In comparison with fBDD and AuNPs/CA/fBDD electrode, the lower semicircle diameter of electrochemical impedance spectroscopy obtained on nBDD and AuNPs/CA/nBDD electrode indicated that the presence of nanograss array improved the reactive site, reduced the interfacial resistance, and made the electron transfer easier. Using electroactive daunomycin as an indicator, the hybridization detection was measured by differential pulse voltammetry.
The experimental results demonstrated that the prepared AuNPs/CA/nBDD electrode was suitable for DNA hybridization with favorable performance of faster response, higher sensitivity, lower detection limit and satisfactory selectivity, reproducibility and stability.
The development of DNA biosensor has attracted considerable attention due to their potential applications, including gene analysis, clinical diagnostics, forensic study and more medical applications [1, 2]. Among the various methods for DNA detection, electrochemical techniques offer great advantages with simplicity, rapidness, relatively low cost, and high sensitivity, and are suitable for the development of inexpensive and portable devices [3–5].
The immobilization of DNA onto the electrodes plays an important role in the fabrication of DNA electrochemical biosensors. With the properties of high surface-to-volume ratio, unique catalytic activity, high electron transfer rate, and strong adsorption ability, gold (Au) nanomaterials with different morphology have been introduced onto the electrode surface through the Au-thiol binding [6–10] or direct immobilization [11–15] for the loading of DNA.
On the other hand, with the advantages of wide potential window, low background current, high stability and excellent resistance to electrode fouling, boron-doped diamond (BDD) film (fBDD) electrode has been used widely for electrochemical detection. In order to add advanced functions, BDD nanograss array electrode (nBDD) was prepared in our previous work .
In the present work, electrochemical DNA biosensor was developed based on the fBDD and the nBDD electrode. The obtained biosensors were characterized by electrochemical impedance spectroscopy (EIS). Probe DNA with thiol group at 5’ end was covalently linked onto the electrode surface through the Au-thiol binding. Hybridization experiment was conducted by immersing the electrode with probe DNA into the hybridization buffer solution containing its complementary sequence. Using electroactive daunomycin (DNR) as an indicator, the hybridization detection was measured by differential pulse voltammetry (DPV).
Chemicals and apparatus
The oligonucleotides were chosen according to the literature  and purchased from TaKaRa Biotechnology Co., Ltd. (Dalian, China).; their base sequences are as follows:
24-base probe sequence (probe DNA): 5′-SH-GAG CGG CGC AAC ATT TCA GGT CGA-3′
its fully complementary sequence: 5′-TCG ACC TGA AAT GTT GCG CCG CTC-3′
a non-complementary sequence: 5′-GAG CGG CGC AAC ATT TCA GGT CGA-3′
Others sequences as following:
5′-TCG ACC TGA AAC GTT GCG CCG CTC-3′ (one-base mismatch);
5′-TCG ACC TCA AAT GTT GAG CCG CTC-3′ (two-base mismatch);
5′-TCG TCC TGA AAC GTT GCG CCT CTC-3′ (three-base mismatch).
Daunomycin hydrochloride was obtained from Shanghai Institute for Drug Control and used without further purification. 10 × TE buffer (0.1 M Tris–HCl, 0.01 M EDTA, pH 8.0) and wash buffer (PBS, 0.01 M phosphate sodium buffer solution, pH 7.4, 0.1 M NaCl) from Shanghai Your Sun Biological Technology Co., Ltd. were all used as-received. Hybridization buffer (0.75 M NaCl, 0.15 M sodium citrate, pH 7.4) was self-prepared. All other reagents were of analytical grade and Millipore Milli-Q ultrapure water was used throughout.
Electrochemical measurements were performed on a CHI660c electrochemical workstation with a three-electrode electrochemical cell. A Ag/AgCl (saturated KCl) reference electrode and a Pt wire counter electrode were used.
Preparation of the AuNPs/CA/fBDD and AuNPs/CA/nBDD electrode
Hybridization and electrochemical detection
Scheme 1 outlined the procedure of DNA hybridization and electrochemical detection on the prepared electrodes. The concrete experimental processes were as follows:
Probe DNA with thiol group at 5’ end was covalently linked onto the electrode surface through the Au-thiol binding, and the processes were as follows: The prepared electrode was immersed into a 1.58 × 10−5 M probe DNA soultion for 12 h with stirring at room temperature. Then the electrode was washed using wash buffer to remove the unimmobilized probe DNA.
The probe DNA-modified electrode was immersed into hybridization buffer solution containing different concentration of target DNA for 30 min at 37°C. After that, the electrode was soaked in wash buffer for removing the non-hybridized target DNA. Then, the hybridized electrode was immersed into the stirred DNR solution (1.0 × 10−5 M) containing PBS (0.1 M, pH 7.4) for 15 min at room temperature. After accumulation of DNR, the electrode was rinsed in PBS (0.1 M, pH 7.4) for 30 s to remove the physically adsorbed molecules . Using the hybridized electrode with accumulated DNR as working electrode, the hybridization detection was measured by DPV in PBS soultion (0.1 M, pH 7.4).
Results and discussion
Characterization of the AuNPs/CA/fBDD and AuNPs/CA/nBDD electrode
Detection of DNA hybridization on the AuNPs/CA/fBDD and AuNPs/CA/nBDD electrode
The immobilization of ssDNA probe onto the electrodes resulted in obvious decreases in DPV responses of DNR (b and b’), which was attributed to the less DNR molecules accessibility to ssDNA probe on the electrode surface . The results also showed that the ssDNA probe had been successfully immobilized on the two electrodes.
After hybridization and DNR intercalation, DPV responses of the hybridized electrodes with accumulated DNR in PBS (0.1 M, pH 7.4) (c and c’) were depicted. As shown in Figure 3c and 3c’, the pronounced increase in the peak current of the DNR located respectively at about −0.69 V and about −0.60 V was observed when hybridized with its complementary sequence. This was ascribed that more DNR could intercalate in the DNA double helix after hybridization, which gave an increased electrochemical response.
In comparison with the AuNPs/CA/fBDD electrode, on the AuNPs/CA/nBDD electrode, the positive shift of peak potential and the obvious increase of all DPV responses demonstrated that the AuNPs/CA/nBDD electrode had advantages with faster response and higher sensitivity and was superior to the AuNPs/CA/fBDD electrode for DNA hybridization.
Selectivity and calibration curve of DNA hybridization on the AuNPs/CA/nBDD electrode
Reproducibility and stability of the DNA biosensor
A series of 8 independent measurements of 1.58 × 10−7 M target DNA were used for estimating the precision, and the relative standard deviation (RSD) was 5.63%. It showed that the proposed DNA biosensor had acceptable reproducibility. The probe DNA modified electrode was first stored in the refrigerator at 4°C over three weeks and then examined via DPV after its hybridization with complementary target DNA. Experiments demonstrated that the DNA biosensor retained about 91% of its initial response.
An electrochemical DNA biosensor was developed based on the AuNPs/CA/nBDD electrode. Both the lower semicircle diameter of EIS and the favorable results of hybridization detection obtained on the AuNPs/CA/nBDD electrode indicated that the presence of nanograss array could improve the reactive site, increase the efficient area of electrode, reduce the interfacial resistance, and make the electron transfer easier. The prepared AuNPs/CA/nBDD electrode was suitable for DNA hybridization with favorable performance of faster response and higher sensitivity, lower detection limit and satisfactory selectivity, reproducibility and stability.
The authors thank the financial support of the National Natural Science Foundation of China (Grant No. 21105022), and the Doctor Foundation of Henan University of Technology (2010BS019).
- Ting BP, Zhang J, Gao Z, Ying JY: A DNA biosensor based on the detection of doxorubicin-conjugated Ag nanoparticle labels using solid-state voltammetry. Biosens Bioelectron. 2009, 25: 282-287. 10.1016/j.bios.2009.07.005.View ArticleGoogle Scholar
- Steichen M, Buess-Herman C: Electrochemical detection of the immobilization and hybridization of unlabeled linear and hairpin DNA on gold. Electrochem Commun. 2005, 7: 416-420. 10.1016/j.elecom.2005.02.015.View ArticleGoogle Scholar
- Kim E, Kim K, Yang H, Kim YT, Kwak J: Enzyme-amplified electrochemical detection of DNA using electrocatalysis of ferrocenyl-tethered dendrimer. Analytiacal Chemistry. 2003, 75: 5665-5672. 10.1021/ac034253x.View ArticleGoogle Scholar
- Wang J, Li J, Baca AJ, Hu J, Zhou F, Yan W, Pang DW: Amplified voltammetric detection of DNA hybridization via oxidation of ferrocene caps on gold nanoparticle/streptavidin conjugates. Analytiacal Chemistry. 2003, 75: 3941-3945. 10.1021/ac0344079.View ArticleGoogle Scholar
- Li C, Liu S, Guo LH, Chen D: A new chemically amplified electrochemical system for DNA detection in solution. Electrochem Commun. 2005, 7: 23-28. 10.1016/j.elecom.2004.10.011.View ArticleGoogle Scholar
- Ding Y, Chen M, Erlebacher J: Mesoporous nanocomposites for electrocatalysis. J Am Chem Soc. 2004, 126: 6876-6877. 10.1021/ja0320119.View ArticleGoogle Scholar
- Li F, Feng Y, Dong P, Yang L, Tang B: Gold nanoparticles modified electrode via simple electrografting of in situgenerated mercaptophenyl diazonium cations for development of DNA electrochemical biosensor. Biosens Bioelectron. 2011, 26: 1947-1952. 10.1016/j.bios.2010.07.076.View ArticleGoogle Scholar
- Pingarrón JM, Yáñez-Sedeño P, González-Cortés A: Gold nanoparticle-based electrochemical biosensors. Electrochim Acta. 2008, 53: 5848-5866. 10.1016/j.electacta.2008.03.005.View ArticleGoogle Scholar
- Lu W, Lin L, Jiang L: Nanogold hollow balls with dendritic surface for hybridization of DNA. Biosens Bioelectron. 2007, 22: 1101-1105. 10.1016/j.bios.2006.03.024.View ArticleGoogle Scholar
- Liu S, Liu J, Han X, Cui Y, Wang W: Electrochemical DNA biosensor fabrication with hollow gold nanospheres modified electrode and its enhancement in DNA immobilization and hybridization. Biosens Bioelectron. 2010, 25: 1640-1645. 10.1016/j.bios.2009.11.026.View ArticleGoogle Scholar
- Sapsford KE, Park D, Goldman ER, Foos EE, Trammell SA, Lowy DA, Ancona MG: Selective DNA-mediated assembly of gold nanoparticles on electroded substrates. Langmuir. 2008, 24: 10245-10252. 10.1021/la800640u.View ArticleGoogle Scholar
- Wang X, Yang T, Li X, Jiao K: Three-step electrodeposition synthesis of self-doped polyaniline nanofiber-supported flower-like Au microspheres for high-performance biosensing of DNA hybridization recognition. Biosens Bioelectron. 2011, 26: 2953-2959. 10.1016/j.bios.2010.11.045.View ArticleGoogle Scholar
- Li F, Han X, Liu S: Development of an electrochemical DNA biosensor with a high sensitivity of fM by dendritic gold nanostructure modified electrode. Biosens Bioelectron. 2011, 26: 2619-2625. 10.1016/j.bios.2010.11.020.View ArticleGoogle Scholar
- Hu K, Lan D, Li X, Zhang S: Electrochemical DNA biosensor based on nanoporous gold electrode and multifunctional encoded DNA-Au Bio bar codes. Anal Chem. 2008, 80: 9124-9130. 10.1021/ac8017197.View ArticleGoogle Scholar
- Hong C, Zhu N, Jiang Y, He P, Fang Y: Cu@Au alloy nanoparticle as oligonucleotides labels for electrochemical stripping detection of DNA hybridization. Biosens Bioelectron. 2003, 18: 1311-1319. 10.1016/S0956-5663(03)00084-8.View ArticleGoogle Scholar
- Wei M, Terashima C, Lv M, Fujishima A, Gu Z-Z: Boron-doped diamond nanograss array for electrochemical sensors. Chem Commun. 2009, 24: 3624-3626.View ArticleGoogle Scholar
- Frens G: Controlled nucleation for the regulation of the particle size in monodisperse gold solutions. Nature (Phys Sci). 1973, 241: 20-25.View ArticleGoogle Scholar
- Hu GZ, Zhang DP, Wu WL, Yang ZS: Selective determination of dopamine in the presence of high concentration of ascorbic acid using nano-Au self-assembly glassy carbon electrode. Colloids Surf B Biointerfaces. 2008, 62: 199-205. 10.1016/j.colsurfb.2007.10.001.View ArticleGoogle Scholar
- Yang Y, Wang Z, Yang M, Li J, Zheng F, Shen G, Yu R: Electrical detection of deoxyribonucleic acid hybridization based on carbon-nanotubes/nano zirconium dioxide/chitosan-modified electrodes. Anal Chim Acta. 2007, 584: 268-274. 10.1016/j.aca.2006.11.055.View ArticleGoogle Scholar
- Tian R-H, Zhi J-F: Fabrication and electrochemical properties of boron-doped diamond film-gold nanoparticle array hybrid electrode. Electrochem Commun. 2007, 9: 1120-1126. 10.1016/j.elecom.2006.12.027.View ArticleGoogle Scholar
- Zhu N, Gu Y, Chang Z, He P, Fang Y: PAMAM dendrimers-based DNA biosensors for electrochemical detection of DNA hybridization. Electroanalysis. 2006, 18: 2107-2114. 10.1002/elan.200603589.View ArticleGoogle Scholar
- Cai H, Cao X, Jiang Y, He P, Fang Y: Carbon nanotube-enhanced electrochemical DNA biosensor for DNA hybridization detection. Anal Bioanal Chem. 2003, 375: 287-293.Google Scholar
This article is published under license to BioMed Central Ltd. This is an Open Access article distributed under the terms of the Creative Commons Attribution License (http://creativecommons.org/licenses/by/2.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.