Electrochemical determination of hydroquinone using hydrophobic ionic liquid-type carbon paste electrodes
© Liu and He 2010
Received: 10 September 2010
Accepted: 26 October 2010
Published: 26 October 2010
Three types of carbon paste electrodes (CPEs) with different liquid binders were fabricated, and their electrochemical behavior was characterized via a potassium hexacyanoferrate(II) probe. 1-Octyl-3-methylimidazolium hexafluorophosphate ionic liquid (IL) as a hydrophobic conductive pasting binder showed better electrochemical performance compared with the commonly employed binder. The IL-contained CPEs demonstrated excellent electroactivity for oxidation of hydroquinone. A diffusion control mechanism was confirmed and the diffusion coefficient (D) of 5.05 × 10-4 cm2 s-1 was obtained. The hydrophobic IL-CPE is promising for the determination of hydroquinone in terms of high sensitivity, easy operation, and good durability.
Hydroquinone (H2Q) is widely used in many fields such as reducing agent, antioxidant, polymerization inhibitor, black white film developer, anthraquinone dye, azo dyestuff and other chemical intermediate[1, 2]. However, because of its toxicity to humans and difficulty in degradation, H2Q is also considered as a serious environmental pollutant. Thus, it is very necessary to find simple and efficient analytical approaches for the determination of H2Q. Several analytical methods have been developed including high performance liquid chromatography[4, 5], flow injection analysis, spectrophotometry[7–9], and electrochemistry[10, 11] etc. In consideration of its good electrochemical activity, electro-oxidation of H2Q could be a competitive method for the determination of H2Q via recording electrochemical response[12–14].
The carbon paste electrode (CPE), first introduced by Adams, is one of the most commonly used electrodes. Due to its low cost, ease of fabrication, high sensitivity and renewable surface, the CPE has been widely applied in the electroanalytical community. A traditional carbon paste electrode is composed of graphite powder and pasting liquids. Typical parameters required for pasting liquids are: i) chemical inertness and electroinactivity, ii) high viscosity and low volatility, and iii) minimal solubility in aqueous solutions. Thus, the commonly used pasting liquids for the preparation of carbon paste were focused on organic mineral oils (e.g. nujol, paraffin)[17, 18]. However, there exist two obvious disadvantages for mineral oil-type binders (or pasting liquids) since they are not component-fixed organic compounds. One is the presence of impurities that may give rise to undesirable effects on trace electroanalysis. The other is the inherent non-conductivity that can diminish the electrochemical sensitivity of the paste electrode. Therefore, the use of alternative viscous liquids with chemical inertness and inherent conductivity as a pasting binder may be very helpful for improving the electroanalytical performance of the CPE.
Ionic liquids (ILs) or room temperature ionic liquids (RTILs) that are entirely ion-composed fluids at ambient temperature have been received much attention in many fields over the recent years[19–28]. In view of high chemical stability, wide electrochemical window, good ionic conductivity, no detectable volatility and suitable viscosity for binding carbon particles, many types of ILs could be served as promising pasting liquids for the preparation of carbon paste. In 2005, we first proposed the use of ILs as liquid binder fabricating a novel IL-type CPE and demonstrated its excellent electrochemical performance in catalysis and detection of electroactive substance19. Much to our gratification, this IL-type CPE has soon attracted a great attention in the fields of electrochemistry and electroanalysis. Moreover, this pioneering work was recognized by the world-renown electroanalysts[29, 30] and followed by lots of colleagues in different teams. Maleki et al. fabricated the IL-type CPE which was found to be electrocatalytic towards the detection of biomolecules such as NADH, dopamine. Shul et al. investigated ion transfer between the electrolyte and the IL-type CPE. Sun et al. fabricated a IL-type CPE with molecular nanofilm of CaCO3 (entrapped in Nafion) as effective sensor for hemoglobin (Hb) via the transformation of Hb-Fe(III)/Fe(II) redox couple. Compton and colleagues demonstrated that the problem of large capacitive currents at IL-type CPE can be overcome by using steady state linear sweep voltammetry at a rotating disk IL-type CPE electrode. Musameh and Wang fabricated an IL-type CPE microelectrode to minimize capacitive currents that further enhanced the faradic response. To the best of our knowledge, there are a lot of interesting programs relevant to IL-type CPEs being underway.
Potassium hexafluorophosphate (analytical grade, Alfa Aesar, USA), hydroquinone (analytical grade, Southern Chemical Reagent Inc., China), mineral oil (chemical purity, changsha Fenlukon Plastics Chemical Plant, China), potassium hexacyanoferrate(II) (analytical grade, Xi'an Chemical Reagent plant, China), heptane (analytical grade, Shantou Guanghua Chemical Plant, China), sodium dihydrogen phosphate, disodium hydrogen phosphate (analytical grade, both from Tianjing 6th Chemical Reagent Plant, China), graphite powder, acetonitrile (Both from Sinopharm Chemical Reagent Co. Ltd, China), and ethyl acetate (analytical grade, Nanjing Chemical Reagent Co. Ltd, China) were used without further treatment. N-methylimidazole (analytical grade, Alfa Aesar, USA) and 1-Bromooctane (chemical purity, Sinopharm Chemical Reagent Co. Ltd, China) were purified as described elsewhere. All aqueous solutions were prepared by using doubly deionized water. Phosphate buffer solution (PBS) was prepared by mixing 0.05 mol L-1 NaH2PO4 and 0.05 mol L-1 Na2HPO4 with 1:1 volume, the pH was then adjusted to 7.0 with 0.1 mol L-1 H3PO4 and 0.1 mol L-1 NaOH.
Synthesis of OMIM-PF6ionic liquids
The imidazolium bromide salts were synthesized with a modified procedure reported elsewhere[36, 37]. Definitely, the freshly distilled N-methylimidazole (4.0 mL, 0.05 mol) and heptane (30 mL) were poured into a 100-mL three-necked flask equipped with a reflux condenser. 1-Bromooctane (8.7 mL, 0.05 mol) was added dropwise through a drop funnel. The mixture solution was stirred at 70°C for 24 h, then cooled to room temperature, a two-layer solution formed. The lower layer was kept and washed repeatedly with ethyl acetate. After distilling, the imidazolium bromide salts were obtained.
The OMIM-PF6 ionic liquids were synthesized with the imidazolium bromide salts and KPF6. The 1-octyl-3-methylimidazolium bromide was transferred in a 250 mL round bottom flask, in which an equimolar amount (0.05 mol) of 130 mL KPF6 solution was added dropwise. The reaction was kept for 24 h at room temperature under stirring. The water-insoluble OMIM-PF6 formed in the bottom of the flask. The resultant OMIM-PF6 was washed repeatedly with water until no free Br- could be detected in the aqueous washings by AgNO3. After dry at 80°C in vacuum for 10 hours, the OMIM-PF6 ionic liquids were sealed in store. As-synthesized OMIM-PF6 ionic liquids were analyzed. IR: 3171 and 3125 [ν(C-H) aromatic]; 2929 and 2858 [ν(C-H) aliphatic]; 838 [ν(P-F)]; 1574 and 1467 [ν(C = C)]. The 1H NMR spectrum contains peaks at δ 8.47, 7.30, 7.27, 4.12, 3.90, 1.85, 0.86.
Fabrication of carbon paste electrodes
The carbon paste electrode was generally prepared as follows: 0.20 g graphite powder and 0.05 mL liquid binder were mixed in an agate mortar and ground for 15 minutes to form a uniform paste. The homogeneous paste was packed into a cavity (1 mm diameter) at the end of a Teflon tube. The electrical contact was provided by a copper wire connected to the paste in the inner hole of the tube. Three types of carbon paste electrodes marked M-CPE, IL-CPE and MIL-CPE were fabricated by using different binder, i.e. mineral oil, OMIM-PF6 ionic liquid and mixture of mineral oil/ionic liquid by half volume, respectively. Prior to use, the surfaces of the as-prepared CPEs were carefully smoothed with the weighing paper.
The electrochemical cell was assembled with a conventional three-electrode system including a well-fabricated CPE as the working electrode, a saturated calomel electrode (SCE) as the reference electrode, and a large area Pt disk as the counter electrode. A CHI 660 electrochemical work station (CH Instruments) was employed for the electrochemical measurements. All measurements were carried out at room temperature (21-23°C). In cyclic voltammetry (CV) measurements, the electrochemical windows of Potassium ferrocyanide and H2Q were -0.3V-0.7V and -0.7V-0.8V, respectively. In electrochemical impedance measurements, the applied perturbation amplitude was 0.005V, the frequencies swept from 104 to 0.1Hz.
Results and discussion
the diffusion coefficient D of H2Q at the OMIM-PF6 IL-CPE could be evaluated from the slope. Where n is number of electrons to oxidize one molecule of H2Q, F is Faraday constant, A is area of the IL-CPE electrode (in cm2), C * is initial concentration of H2Q (in mol cm-3), and t is time (in s). The D was calculated as 5.05 × 10-4 cm2 s-1, which was larger than the reported values of 2.26 × 10-5 cm2 s-1 at N-butylpyridinium hexafluorophosphate (BPPF6) CPE13 and 7.88 × 10-5 cm2 s-1 at glassy carbon electrode. This result quantitatively disclosed a good diffusion reaction of H2Q at the OMIM-PF6 IL-CPE.
Three types of CPEs, namely as M-CPE, IL-CPE, and MIL-CPE were fabricated based on different liquid binders including mineral oil, OMIM-PF6, and their mixture with half doping, respectively. The hydrophobic OMIM-PF6 as binder component showed better electrochemical performance of the CPE due to the enhanced charge-transfer capability of the conductive IL. The IL-contained CPEs presented excellent electroactivity for oxidation of H2Q, and the electrochemical oxidation process was controlled by diffusion of H2Q. The diffusion coefficient (D) at the IL-CPE was calculated as 5.05 × 10-4 cm2 s-1 larger than the literature value at BPPF6 CPE or glassy carbon electrode. The hydrophobic IL-CPE with high sensitivity, easy operation, and good durability showed a promising application for the determination of H2Q.
This work was financially supported by the National Natural Science Foundation of China (No.20976198), Hunan Provincial Natural Science Foundation of China (No. 10JJ2004), and the Specialized Research Fund for the Doctoral Program of Higher Education of China (20090162120012).
- Zhao GH, Li MF, Hu ZH, Li HX, Cao TC: Electrocatalytic redox of hydroquinone by two forms Of L-proline. J Mole Cataly A: Chem. 2006, 255 (1-2): 86-91. 10.1016/j.molcata.2006.03.039.View ArticleGoogle Scholar
- Subramanyam R, Mishra IM: Treatment of catechol bearing wastewater in an upflow anaerobic sludge blanket (UASB) reactor: sludge characteristics. Bioresource Technol. 2008, 99 (18): 8917-8925. 10.1016/j.biortech.2008.04.067.View ArticleGoogle Scholar
- Vieira IC, Fatibello-Filho O: Biosensor based on paraffin/graphite modified with sweet potato tissue for the determination of hydroquinone in cosmetic cream in organic phase. Talanta. 2000, 52: 681-689. 10.1016/S0039-9140(00)00420-3.View ArticleGoogle Scholar
- Lin CH, Sheu JY, Wu HL, Huang YL: Determination of hydroquinone in cosmetic emulsion using microdialysis sampling coupled with high-performance liquid chromatography. J Pharm Biomed Anal. 2005, 38: 414-419. 10.1016/j.jpba.2005.01.031.View ArticleGoogle Scholar
- Asan A, Isildak I: Determination of major phenolic compounds in water by reversed-phase liquid chromatography after pre-column derivatization with benzoyl chloride. J Chromatogr A. 2003, 988 (1): 145-149. 10.1016/S0021-9673(02)02056-3.View ArticleGoogle Scholar
- Chen GN, Liu JS, Duan JP, Chen HQ: Coulometric detector based on porous carbon felt working electrode for flow injection analysis. Talanta. 2000, 53: 651-660. 10.1016/S0039-9140(00)00551-8.View ArticleGoogle Scholar
- Gajewska T, Kaszuba A: The UV spectrophotometrica determination of migrated hydroquinone. Polimery. 1999, 44: 690-696.Google Scholar
- Pistonesi MF, Nezio MSD, Centurión ME, Palomeque ME, Lista AG, Band BSF: Determination of phenol, resorcinol and hydroquinone in air samples by synchronous fluorescence using partial least-squares (PLS). Talanta. 2006, 69 (5): 1265-1268. 10.1016/j.talanta.2005.12.050.View ArticleGoogle Scholar
- Fiamegos YC, Stalikas CD, Pilidis GA, Karayannis MI: Synthesis and analytical applications of 4-aminopyrazolone derivatives as chromogenic agents for the spectrophotometric determination of phenols. Anal Chim Acta. 2000, 403 (1-2): 315-323. 10.1016/S0003-2670(99)00644-3.View ArticleGoogle Scholar
- McCreery RL: Electroanalytical Chemistry: A Series of Advances. Edited by: Bard AJ. 1991, Marcel Dekker; New YorkGoogle Scholar
- Wang LH: Simultaneous determination of hydroquinone ethers in cosmetics after preconcentration at a carbon paste electrode. Analyst. 1995, 120: 2241-2244. 10.1039/an9952002241.View ArticleGoogle Scholar
- Zhang Y, Zheng JB: Comparative investigation on electrochemical behavior of hydroquinone at carbon ionic liquid electrode, ionic liquid modified carbon paste electrode and carbon paste electrode. Electrochim Acta. 2007, 52: 7210-7216. 10.1016/j.electacta.2007.05.039.View ArticleGoogle Scholar
- Sun W, Jiang Q, Yang MX, Jiao K: Electrochemical Behaviors of Hydroquinone on a Carbon Paste Electrode with Ionic Liquid as Binder. Bull Korean Chem Soc. 2008, 29 (5): 915-920. 10.5012/bkcs.2008.29.5.915.View ArticleGoogle Scholar
- Li MG, Ni F, Wang YL, Xu SD, Zhang DD, Chen SH, Wang L: Sensitive and Facile Determination of Catechol and Hydroquinone Simultaneously Under Coexistence of Resorcinol with A Zn/Al Layered Double Hydroxide Film Modified Glassy Carbon Electrode. Electroanalysis. 2009, 21 (13): 1521-1526. 10.1002/elan.200804573.View ArticleGoogle Scholar
- Adams RN: Carbon paste electrodes. Anal Chem. 1958, 30: 1576-10.1021/ac60141a600.View ArticleGoogle Scholar
- Adams RN: Electrochemistry at Solid Electrodes. 1969, Marcel Dekker; New YorkGoogle Scholar
- Tomcik P, Banks CE, Davies TJ, Compton RG: A Self-Catalytic Carbon Paste Electrode for the Detection of Vitamin B12. Anal Chem. 2004, 76: 161-165. 10.1021/ac030308j.View ArticleGoogle Scholar
- Almeida CMVB, Giannetti BF: A new and practical carbon paste electrode for insoluble and ground samples. Electrochem Commun. 2002, 4: 985-988. 10.1016/S1388-2481(02)00511-8.View ArticleGoogle Scholar
- Liu HT, He P, Li ZY, Sun CY, Shi LH, Liu Y, Zhu GY, Li JH: An ionic liquid-type carbon paste electrode and its polyoxometalate-modified properties. Electrochem Commun. 2005, 7: 1357-1363. 10.1016/j.elecom.2005.09.018.View ArticleGoogle Scholar
- Conway BE: Electrochemical Supercapacitors: Scientific Fundamentals and Technological Applications. 1999, Kluwer Academic/Plenum Publishers; New YorkView ArticleGoogle Scholar
- Zhou Y, Antonietti M: Synthesis of Very Small TiO2 Nanocrystals in a Room-Temperature Ionic Liquid and Their Self-Assembly toward Mesoporous Spherical Aggregates. J Am Chem Soc. 2003, 125 (49): 14960-14961. 10.1021/ja0380998.View ArticleGoogle Scholar
- Ranu BC, Jana R: Ionic Liquid as Catalyst and Reaction Medium: A Simple, Convenient and Green Procedure for the Synthesis of Thioethers, Thioesters and Dithianes using an Inexpensive Ionic Liquid, [PMIM]Br. Adv Syn Catal. 2005, 347 (14): 1811-1818. 10.1002/adsc.200505122.View ArticleGoogle Scholar
- Buzzeo MC, Giovanelli D, Lawrence NS, Hardacre C, Seddon KR, Compton RG: Elucidation of the electrochemical oxidation pathway of ammonia in dimethylformamide and the room temperature ionic liquid, 1-ethyl-3-methylimidazolium bis(trifluoromethylsulfonyl)imide. Electroanalysis. 2004, 16: 888-896. 10.1002/elan.200302910.View ArticleGoogle Scholar
- Liu HT, He P, Li ZY, Liu Y, Li JH: A novel nickel-based mixed rare-earth oxide/activated carbon supercapacitor using room temperature ionic liquid electrolyte. Electrochim Acta. 2006, 51: 1925-1931. 10.1016/j.electacta.2005.06.034.View ArticleGoogle Scholar
- Liu HT, He P, Li ZY, Liu Y, Li J, Zheng LZ, Li JH: The Inherent Capacitive Behavior of Imidazolium-based Room-Temperature Ionic Liquids at Carbon Paste Electrode. Electrochem Solid-State Lett. 2005, 8 (7): J17-J19. 10.1149/1.1922870.View ArticleGoogle Scholar
- Safavi A, Maleki N, Honarasa F, Tajabadi F, Sedaghatpour F: Ionic Liquids Modify the Performance of Carbon Based Potentiometric Sensors. Electroanalysis. 2007, 19 (5): 582-586. 10.1002/elan.200603767.View ArticleGoogle Scholar
- Liu HT, He P, Li ZY, Sun DZ, Huang HP, Li JH, Zhu GY: Crystalline Vanadium Pentoxide with Hierarchical Mesopores and Its Capacitive Behavior. Chem Asian J. 2006, 1: 701-706. 10.1002/asia.200600169.View ArticleGoogle Scholar
- Liu HT, Zhu GY: The electrochemical capacitance of nanoporous carbons in aqueous and ionic liquids. J Power Sources. 2007, 171: 1054-1061. 10.1016/j.jpowsour.2007.06.200.View ArticleGoogle Scholar
- Svancara I, Walcarius A, Kalcher K, Vytras K: Carbon paste electrodes in the new millennium. Cent Eur J Chem. 2009, 7 (4): 598-656. 10.2478/s11532-009-0097-9.Google Scholar
- Švancara I, Vytřas K, Kalcher K, Walcarius A, Wang J: Carbon Paste Electrodes in Facts, Numbers, and Notes: A Review on the Occasion of the 50-Years Jubilee of Carbon Paste in Electrochemistry and Electroanalysis. Electroanalysis. 2009, 21 (1): 7-28. 10.1002/elan.200804340.View ArticleGoogle Scholar
- Maleki N, Safavi A, Tajabadi F: High-Performance Carbon Composite Electrode Based on an Ionic Liquid as a Binder. Anal Chem. 2006, 78: 3820-3826. 10.1021/ac060070+.View ArticleGoogle Scholar
- Shul G, Sirieix-Plenet J, Gaillon L, Opallo M: Ion transfer at carbon paste electrode based on ionic liquid. Electrochem Commun. 2006, 8: 1111-1114. 10.1016/j.elecom.2006.05.002.View ArticleGoogle Scholar
- Sun W, Gao RF, Jiao K: Electrochemistry and Electrocatalysis of Hemoglobin in Nafion/nano-CaCO3 Film on a New Ionic Liquid BPPF6 Modified Carbon Paste Electrode. J Phys Chem B. 2007, 111 (17): 4560-4567. 10.1021/jp067933n.View ArticleGoogle Scholar
- Musameh M, Wang J: Sensitive and stable amperometric measurements at ionic liquid-carbon paste microelectrodes. Anal Chim Acta. 2008, 606: 45-49. 10.1016/j.aca.2007.11.012.View ArticleGoogle Scholar
- Armarego WLF, Chai CLL: Purification of Laboratory Chemicals. 2003, Elsevier Science; New York, 5Google Scholar
- Huddleston JG, Visser AE, Reichert WM, Willauer HD, Broker GA, Rogers RD: Characterization and comparison of hydrophilic and hydrophobic room temperature ionic liquids incorporating the imidazolium cation. Green Chem. 2001, 3 (4): 156-164. 10.1039/b103275p.View ArticleGoogle Scholar
- Paul A, Mandal PK, Samanta A: On the Optical Properties of the Imidazolium Ionic Liquids. J Phys Chem B. 2005, 109: 9148-9153. 10.1021/jp0503967.View ArticleGoogle Scholar
- Zhao F, Wu XE, Wang MK, Liu Y, Gao LX, Dong SJ: Electrochemical and bioelectrochemistry properties of room-temperature ionic liquids and carbon composite materials. Anal Chem. 2004, 76: 4960-4967. 10.1021/ac0494026.View ArticleGoogle Scholar
- Rozniecka E, Shul G, Sirieix-Plenet J, Gaillon L, Opallo M: Electroactive ceramic carbon electrode modified with ionic liquid. Electrochem Commun. 2005, 7 (3): 299-304. 10.1016/j.elecom.2005.01.013.View ArticleGoogle Scholar
- Bard AJ, Faulkner LR: Electrochemical Methods: Fundamentals and Applications. 2001, John Wiley & Sons; New York, 2Google Scholar
- Peng J, Gao ZN: Influence of micelles on the electrochemical behaviors of catechol and hydroquinone and their simultaneous determination. Anal Bioanal Chem. 2006, 384 (7-8): 1525-1532. 10.1007/s00216-006-0329-1.View ArticleGoogle Scholar